Condensers are either air-cooled or water-cooled, or both.
The exhauster is practicall y a rotary gas pump which serves the purpose of drawing the gas from the hydraulic main through the condensers, and then forcing it through the purifying vessels to the holder.
Especially powerful achromatic condensers are really only magnified microscope objectives, with the difference that they are not corrected for the thickness of the cover slip, but for the thickness of the glass on which the object is placed.
Condensers are for this reason also constructed with apertures up to 1 40.
By a modification of the bridge method, applied with excellent results by Dr Muirhead to submarine work, condensers are substituted for a and b, one being also placed in the circuit between P and Q.