Hot-wire ammeters are open to the following objections: The scale divisions for equal increments of current are not equal in length, being generally much closer together in the lower parts of the scale.
- Another large class of ammeters depend for their action upon the fact that an electric current create; an electric field round its conductor, which varies in strength from point to point, but is otherwise proportional to the current.
In a small piece of soft iron, as in the case of the corresponding ammeters, and this in turn may be made to displace an indicating needle over a scale so that corresponding to every given potential difference between the terminals of the instrument there is a corresponding fixed position of the needle on the scale.
Like the corresponding ammeters, they have the great advantage that the scales are equidivisional and that there is no dead part in the scale, whereas both the electrostatic and electrothermal voltmeters, above described, labour under the disadvantage that the scale divisions are not equal but increase with rise of voltages, hence there is generally a portion of the scale near the zero point where the divisions are so close as to be useless for reading purposes and are therefore omitted.
Ammeters to measure the volume, and voltmeters to determine the pressure of current supplied to the baths, should also be provided.