In grating spectroscopes both plane and concave gratings are employed in connexion with a collimator and observing telescope.
Rutherfurd devised one made of flint glass with two crown glass compensating prisms; whilst Thallon employed a hollow prism containing carbon bisulphide also compensated by flint glass prisms. In direct vision spectroscopes the refracting prisms and slit are in the observing telescope.
- T he modern equatorial should, for general purposes, be capable of carrying spectroscopes of considerable weight, so that the proportional strength of the axes and the rigidity of the instrument have to be considerably increased.
But if it be possible to mount a fixed telescope by which a solar or stellar image can be formed within a laboratory we give the following advantages: - (1) There is no mechanical limit to the length of the telescope; (2) the clockwork and other appliances to move the mirror, which reflects the starlight along the axis, are much lighter and smaller than those required to move a large telescope; (3) the observer remains in a fixed position, and spectroscopes of any weight can be used on piers within the laboratory; and (4) the angular value of any linear distance on a photographic plate can be determined by direct measurement of the distance of the photographic plate from the optical centre of the object-glass.